Agilent Single Quadrupole Repair and Operational Instructional Videos
Each video provides detailed, in depth instruction to help the user understand the basic theory and specifics of the Agilent single quadrupole. Addition manuals, links and downloads are available in each subject. It is reccommended you start with the hardware overview and do each video in order.
All videos and content are designed to be used with your PC, smart phone and tablet.
Agilent Single Quad 5975 GCMS Hardware Overview Video. This learning video will cover the major
components of the Mass Spectrometer. These include the E-MOD,
vacuum components and other parts.
Agilent Single Quad 5975 GCMS Source Overview Video. This video will present an overview of the
source including function structure and common problems
Agilent Single Quad 5975 GCMS Detector Overview Video. In this video we will present an overview
of the detector on the Single QUad. This includes the High energy dynode, or H.E.D , the Electron multiplier
or E.M. and the log amplifier. We will also cover the circuitry and test points for trouble shooting
Agilent Single Quad 5975 GCMS Interface Overview Video. In this video we will discuss how the MS interfaces
with the GC. We will disuss the differences between the EI and CI versions of the
transferline including common problems
Agilent Single Quad 5975 GCMS Mass filter Overview Video. In this video we will talk about the functions
of the mass filter and how it works. We will identify quad related problems and adjustments from
the tune report. We will also talk about removing and cleaning the quad.
Agilent Single Quad 5975 GCMS Electronics Overview Video. This video will present an overview of the single quad electronics. We will discuss the main electronic circuit boards and their primary functions. These boards include the Local user interface or Louie, the Smart Card , the Side board, the Main board , Log Amp and power supplies.
Agilent Single Quad 5975 GCMS Tuning Overview Video. In this video we will cover the fundamentals of
MS tuning. This will including the various tuning methods available on the chemstation or mass hunter
software. We will also cover Manual tuning and gain tuning.
Agilent Single Quad 5975 GCMS Acquisition Overview Video presents an introduction to data
acquisition. We will discussacquisition methods and principles, scan and sim principles, E M Saver and
Trace ion detection.
Agilent Single Quad 5975 GCMS CI theory Overview Video will provide a brief overview of the
Positive and Negative CI theory as it relates to the Single Quad
Agilent Single Quad 5975 GCMS Instrument Configuration Overview Video. In this video you will learn how
to configure a GC MS instrument in the chemstation or mass hunter program. This includes updating the
firmware, knowing what files are copied and the general file structure
Agilent Single Quad 5975 GCMS Vacuum Overview Video. In this video we will go over the main types of
pumps used with the mass spectrometer. We will also go over identifying different types of vacuum
problems and how to troubleshoot them.
Agilent Single Quad 5975 GCMS Installation Overview Video. This video will discuss the unique Site Prep and
Installation steps needed when moving or re-installing the 5975 mass spectrometer. These include Gas
Requirements, power considerations, column installation and Ion Gauge installation.
The CI installation video will discuss CI installation and tuning of the single quad. It will first cover a hardware overview including the CI installation kit, then will cover the CI tunes and checkout samples. Lastly, the video covers common problems and solutions
Agilent Single Quad 5975 GCMS Quadrupole Cleaning Video. This video will provide an overview of cleaning
the quad on the mass spectrometer.